|Place of Origin:||Dongguan,China|
|Minimum Order Quantity:||one set|
|Price:||USD499 to USD699|
|Packaging Details:||standard exporting package|
|Delivery Time:||3 days|
|Payment Terms:||L/C, D/A, D/P, T/T, Western Union, MoneyGram|
|Supply Ability:||300 sets per month|
|Application:||Aluminium Oxide||Material:||Stainless Steel|
|Standards:||ISO 903||Testing Before Delivery:||Yes|
|After-sale Customer Service:||Yes||Product Name:||Bulk Density Testing Machine|
ISO 903 Aluminium Oxide Untamped Density Meter , Bulk Density Tester , Bulk Density Apparatus
Comply with GB/T 6609.25-2004, ISO 903:1976 Determination of bulk density Alumina chemical analysis method and determination of physical properties" Alumina - Determination of chemical properties - Determination of bulk density - Part 25
Principle: The sample freely falls from a fixed height without vibration, fills a fixed container of known volume, and calculates the bulk density according to the mass and volume of the sample.
1. Funnel: The inner diameter of the mouth is 100mm, the taper is 60°, and the inside of the discharge port is 6mm.
2. Measuring cylinder: about 200ml, the ratio of inner diameter to inner height is 1:6, base outer diameter is 70mm
3. Material: stainless steel
4. Annular funnel holder: Install a ring frame with a fixed funnel on the sturdy base with pillars. The long screws fix the ring frame on the pillars and can be freely adjusted. The funnel is mounted above the predetermined level of the cylindrical container.