|Place of Origin:||Dongguan,China|
|Minimum Order Quantity:||one set|
|Packaging Details:||standard exporting package|
|Delivery Time:||3 days|
|Payment Terms:||L/C, D/A, D/P, T/T, Western Union, MoneyGram|
|Supply Ability:||300 sets per month|
|Application:||Plastic Materials||The Height Of The Measuring Cup:||24±0.1cm|
|The Height Of The Funnel And The Measuring Cup:||3.8cm||Testing Before Delivery:||Yes|
|After-sale Customer Service:||Yes||Product Name:||Bulk Density Testing Equipment|
powder flow tester,
powder flow analyzer
ASTM D-1895-B Apparent Density Tester , Bulk Density Meter Method B for Plastic
1. The height of the measuring cup is 24±0.1cm;
2. The height of the funnel and the measuring cup is 3.8cm;
3. The outlet of the funnel is 2.54cm±0.16cm;
4. The opening of the funnel is 12.7±1.0cm;
5. The volume of the measuring cup is 400ml±2.0ml.;
6. Angle: 24 degrees;
7.Material: stainless steel;
8.Configuration accessories: 1 pcs of brush;1 pcs of 15cm steel ruler;
Structure and operation method
(1)Structure: The analyzer consists of a base, a support frame, a funnel, and a receiver.
Receiver volume 400ML±0.5ML
a.Open the package, take out the analyzer, and check the parts according to the packing list.
b.Place the analyzer on a stable work surface.
c.Block the lower outlet of the stainless steel funnel with any flat material, and place the receiver under the funnel, coaxial with the funnel.
d.Take a sample of 410ML-420ML from the measuring cylinder and slowly place it in the funnel.
e.Quickly remove the insert, so that all the samples are naturally leaked (the receiver that is filled with the sample must not be shaken).
f.Use a ruler to scrape off the excess sample on the mouth of the receiver, and then weigh it on the drug balance to the nearest 0.2g (the sample must not be repeatedly measured).
5. Packing List
|Steel ruler:||150mm 1 pcs|
|Receiving container:||400ml 1 pcs|
Contact Person: Mrs. Julie Zhong