|Place of Origin:||dongguan|
|Minimum Order Quantity:||1|
|Packaging Details:||42 X 16 X 32cm|
|Delivery Time:||2 days for K Value of Precious Metals Analyzer|
|Payment Terms:||L/C, D/A, D/P, T/T, Western Union, MoneyGram,other|
|Supply Ability:||300 Set/Sets per Month|
|Power Supply:||AC 100V - 240V||OEM Service:||Yes|
|Density Resolution:||0.001 G/cm3||Weight Resolution:||0.005g|
|Usage:||Gold Testing Machine||Warranty:||2 Years;life-long Maintenance|
precious metal analyzer machine,
electronic precious metal tester
300g DahoMeter Digital Portable Gold Content Tester , Electronic Densimeter Gold Karat Testing Machine
pawn broking, bank, jewelry shop, precious-metals recovery, jewelry factory, Gemstone Testing Center,precious metals researching.etc.
In accordance with archimedes principle water buoyancy method, GB/T1423 of precious metals testing standard,and so on.
Combine archimedes principle with micro electronic technique;adopt sensor imported from Germany,the purity can be calculated through measuring density of precious metals.features of no damage on sample;simple operation,fast and accurate.
K value of gold,PT value of platinum,density value,purity percentage can be measured.
Necessary weight for density:4g
a) Place sample on testing board to measure weight in air,press ENTER key to memory.
b) Place sample into water to measure the weight in water,press ENTER key.the K value of gold,purity percent,density value will be displayed.
Host,sink,test board,tweezer,thermometer,weight,wind and dust cover,a set of measuring particle parts,a set of measuring floating body parts,power transformer
|Testing board||Completed high-capacity plastic testing board|
|Measuring type||Traditional gold and platinum jewelry,necklace,bracelet,ring,large bullion,craft,and gold,silver,copper,cadmium,nickel,tin.etc.|
Display k value
The K value of mixture of gold,silver,and copper;
The K value of mixture of gold and copper;
The K value of mixture of gold and silver;
Display purity percent
The purity percentage of mixture of gold,silver and copper; ;
The purity percentage of mixture of gold and copper;
The purity percentage of mixture of gold and silver;
|K value range||9K~24K|
|Copper and silver proportion setting||Cycle seting(default 7:3)|
|Display value||PT percent,density value,volume|
|PT value range||Pt600 - Pt1000|
|Other Precious Metals||Function||Only for DA series and DE series:measure the purity percentage of main materials in two kinds of mixture|
|Display value||Only for DA series and DE series:density value,volume,content precentage|
|setting||Only for DA series and DE series:density value of main materials and secondary materials setting|
completed injection forming transparent professional measuring tank
|Parameter setting||water temperature setting,measuring medium density setting|
|Memory method||Two procedures|
|Correction method||Auto calibration with one key,auto detection.|
|Output method||Output and print testing data conveniently with standard communication interface of RS-232C.|
|Standard accessories||Host,tank,test board,tweezer,thermometer,weight,wind and dust cover,a set of measuring particle parts,a set of measuring floating body parts,power transformer|
|wind and dust cover||Adopt the unicase dedicated wind and dust cover|
|Option accessories||DE-40 printer|
|power||AC 100V – 240V(Europe standard)|
|Main features||Simple operation,digital display,accurate measuring,stable,durable.|
Our machine has the interface of RS-232C.which can be connected with computer or printer.
For connecting computer,there is no need to install extra driver or software in order to output data from density meter to computer.the driver is including into the windows system when install initially,all need to do is setup one parameter (which is Baud rate) of computer to 9600.Then all is done.
Operation like this:(windows system)
Control panel------>Accessibility options------->general option------>serial key devices (check)------>serial port:COM1(check);baud rate:9600(check)------>done
Parameter setting for the gold testing machine:
Long press Zero + C key at the same time,the screen will be displaying AUT 1,If not,press B key to shift it to AUT 1,Press A key to change AUT 1 to AUT 0,press ENTER to confirm.
After connect to computer.need to creat one new excel.open the excel,choose one blank space.
When the gold testing machine do the testing .the testing result will be displayed at the blank space automatically.
Contact Person: Mrs. Julie Zhong