|Place of Origin:||dongguan|
|Minimum Order Quantity:||1|
|Packaging Details:||wood standard exported package|
|Delivery Time:||5 days|
|Payment Terms:||L/C, D/A, D/P, T/T, Western Union, MoneyGram,other|
|Supply Ability:||300 Set/Sets per Monter|
|Max. Load:||500N||Effective Travel:||200mm|
|Test Speed:||25-280 Mm/min(choose By Customer)||After-sale Customer Service:||Yes|
|Usage:||Push And Pull Force Test Machine||Test Before Delivery:||Yes|
Electric Push Pull Force Test Vertical Machine / Tensile and Pressure Test Machine
AEL Single-column Vertical Machine is force load test platform specially matched for HF and NK series force measurement gauge. The product is double vertical rod structure, featuring good stability, wide applicable scope and convenience, as well as stepless speed regulation, manual (point) and automatic control switching functions. Applicable for tests of force load, Insertion and withdrawal, destruction in such industries as rubber and plastic, light industry-textile, constructional door and window, composite materials, electric wire and cable, parts and accessories for automobile, power machinery, scientific research institutes, etc.
1,Electric load, stepless speed regulation, gear drive
2,Compact structure, stable drive
3,Simple operation, able to be used with various kinds of clips
Ⅲ Specifications and Parameters
Ⅳ Installation Method of Force Measurement Gauge
Sensor in-built type: insert 4-M6 screw into fixed block 4-φ6.2 holes, invaginate 1 gasket respectively in the direction of thread, and screw tight with 4-M6 thread on the installation board of force measurement gauge, the installation holes of force measurement gauge on installation board are divided into two groups, 30x145 group and 40x90 group, which can be determined by users according to requirements.
Ⅴ Usage of Control Panel
1. Power switch: when button is pressed, the power is on, when button is pressed again, the power is off.
2. Speed control: due to the difference of the tested objects and user requirements, the movement speed is 20～200mm/min when turn button of speed regulator is adjusted.
Ⅵ Function Control
Manual/Automation. When turn button is turned to Manual, if carriers are needed to be moved, lift or lower key shall be pressed, if the key is off, the movement stop. When the turn button is turned to Automation, if carriers are needed to be moved, lift or lower key shall be directly pressed.
Ⅶ Travel Adjustment
There are two screws on the back side of the vertical column, loose the screw by standard Hex Wrench and adjust to proper position, and control slide block, then requirement is met.
Ⅷ Usage of Counter
The upper of counter is digital display for recording test times, there are 4 pre-set buttons in the lower part, they are single-digit, ten-digit, hundred-digit and thousand-digit, which mean the pre-set scope is within 0~9999. E.g.: user continuously test some product, the test times is 9999, therefore it is needed to adjust the 4 buttons to display 9999, and press Manual/Automation button, thus 9999 times can be tested, then they stop automatically after the 9999 times is accomplished. If the test is necessary to be implemented, press Stop button.